The human eye can distinguish as many as 10,000 different colours but is far less sensitive to variations in intensity1, meaning that colour is highly desirable when interpreting images. However, most biological samples are essentially transparent, and nearly invisible when viewed using a standard optical microscope2. It is therefore highly desirable to be able to produce coloured images without needing to add any stains or dyes, which can alter the sample properties. Here we demonstrate that colorimetric histology images can be generated using full-sized plasmonically active microscope slides. These slides translate subtle changes in the dielectric constant into striking colour contrast when samples are placed upon them. We demonstrate the biomedical potential of this technique, which we term histoplasmonics, by distinguishing neoplastic cells from normal breast epithelium during the earliest stages of tumorigenesis in the mouse MMTV-PyMT mammary tumour model. We then apply this method to human diagnostic tissue and validate its utility in distinguishing normal epithelium, usual ductal hyperplasia, and early-stage breast cancer (ductal carcinoma in situ). The colorimetric output of the image pixels is compared to conventional histopathology. The results we report here support the hypothesis that histoplasmonics can be used as a novel alternative or adjunct to general staining. The widespread availability of this technique and its incorporation into standard laboratory workflows may prove transformative for applications extending well beyond tissue diagnostics. This work also highlights opportunities for improvements to digital pathology that have yet to be explored.
Subscription info for Chinese customers
We have a dedicated website for our Chinese customers. Please go to naturechina.com to subscribe to this journal.
Rent or Buy article
Get time limited or full article access on ReadCube.
All prices are NET prices.
All data sets are included in this article and its Supplementary Information files. Pathology scoring for Figs. 3d and 4e was conducted on the basis of visual inspection of the slides by expert breast cancer pathologists using an Olympus BX51 optical microscope. Histology images used for this study are available in figshare with the identifier https://doi.org/10.6084/m9.figshare.14897697.v2. Source data are provided with this paper.
Judd, D. B. & Wyszecki, G. Color in Business, Science, and Industry (Wiley & Sons, 1975).
Alberts, B. Molecular Biology of the Cell (Garland Science, Taylor & Francis, 2008).
Fu, Y., Tippets, C. A., Donev, E. U. & Lopez, R. Structural colors: from natural to artificial systems. Wiley Interdiscipl. Rev. Nanomed. Nanobiotechnol. 8, 758–775 (2016).
Lochbihler, H. Colored images generated by metallic sub-wavelength gratings. Opt. Express 17, 12189–12196 (2009).
Lee, H.-S., Yoon, Y.-T., Lee, S.-S., Kim, S.-H. & Lee, K.-D. Color filter based on a subwavelength patterned metal grating. Opt. Express 15, 15457–15463 (2007).
Inoue, D. et al. Polarization independent visible color filter comprising an aluminum film with surface-plasmon enhanced transmission through a subwavelength array of holes. Appl. Phys. Lett. 98, 093113 (2011).
Duempelmann, L., Casari, D., Luu-Dinh, A., Gallinet, B. & Novotny, L. Color rendering plasmonic aluminum substrates with angular symmetry breaking. ACS Nano 9, 12383–12391 (2015).
Raj Shrestha, V., Lee, S.-S., Kim, E.-S. & Choi, D.-Y. Polarization-tuned dynamic color filters incorporating a dielectric-loaded aluminum nanowire array. Sci. Rep. 5, 12450 (2015).
Liddle, J. A. & Gallatin, G. M. Nanomanufacturing: a perspective. ACS Nano 10, 2995–3014 (2016).
Vala, M. & Homola, J. Flexible method based on four-beam interference lithography for fabrication of large areas of perfectly periodic plasmonic arrays. Opt. Express 22, 18778–18789 (2014).
Wang, L. et al. Large area plasmonic color palettes with expanded gamut using colloidal self-assembly. ACS Photon. 3, 627–633 (2016).
Franklin, D. et al. Polarization-independent actively tunable colour generation on imprinted plasmonic surfaces. Nat. Commun. 6, 7337 (2015).
Solak, H. H., Dais, C. & Clube, F. Displacement Talbot lithography: a new method for high-resolution patterning of large areas. Opt. Express 19, 10686–10691 (2011).
Choudhury, K. R., Yagle, K. J., Swanson, P. E., Krohn, K. A. & Rajendran, J. G. A robust automated measure of average antibody staining in immunohistochemistry images. J. Histochem. Cytochem. 58, 95–107 (2010).
Wong, T. T. W. et al. Fast label-free multilayered histology-like imaging of human breast cancer by photoacoustic microscopy. Sci. Adv. 3, e1602168 (2017).
Fereidouni, F. et al. Microscopy with ultraviolet surface excitation for rapid slide-free histology. Nat. Biomed. Eng. 1, 957–966 (2017).
Neilson, K. A. et al. Less label, more free: approaches in label-free quantitative mass spectrometry. Proteomics 11, 535–553 (2011).
Lu, F.-K. et al. Label-free DNA imaging in vivo with stimulated Raman scattering microscopy. Proc. Natl Acad. Sci. USA 112, 11624–11629 (2015).
Ji, M. et al. Rapid, label-free detection of brain tumors with stimulated Raman scattering microscopy. Science Transl. Med. 5, 201ra119 (2013).
Lee, M. et al. Label-free optical quantification of structural alterations in Alzheimer’s disease. Sci. Rep. 6, 31034 (2016).
Langley, D. P., Balaur, E., Hwang, Y., Sadatnajafi, C. & Abbey, B. Optical chemical barcoding based on polarization controlled plasmonic nanopixels. Adv. Funct. Mater. 28, 1704842 (2018).
Langley, D., Balaur, E., Sadatnajafi, C. & Abbey, B. Dual pitch plasmonic devices for polarization enhanced colour based sensing. Proc. SPIE 10013, 1001338 (2016).
Huth, M. et al. Focused electron beam induced deposition: a perspective. Beilstein J. Nanotechnol. 3, 597–619 (2012).
Ding, X. et al. Surface plasmon resonance enhanced light absorption and photothermal therapy in the second near-infrared window. J. Am. Chem. Soc. 136, 15684–15693 (2014).
Kawata, S., Ichimura, T., Taguchi, A. & Kumamoto, Y. Nano-Raman scattering microscopy: resolution and enhancement. Chem. Rev. 117, 4983–5001 (2017).
Duivenvoorden, H. M., Spurling, A., O’Toole, S. A. & Parker, B. S. Discriminating the earliest stages of mammary carcinoma using myoepithelial and proliferative markers. PLoS ONE 13, e0201370 (2018).
Wang, Z., Tangella, K., Balla, A. & Popescu, G. Tissue refractive index as marker of disease. J. Biomed. Opt. 16, 116017 (2011).
Shan, M., Kandel, M. E. & Popescu, G. Refractive index variance of cells and tissues measured by quantitative phase imaging. Opt. Express 25, 1573–1581 (2017).
Giannios, P. et al. Visible to near-infrared refractive properties of freshly-excised human-liver tissues: marking hepatic malignancies. Sci. Rep. 6, 27910 (2016).
Shi, P. et al. Automated Ki-67 quantification of immunohistochemical staining image of human nasopharyngeal carcinoma xenografts. Sci. Rep. 6, 32127 (2016).
van Seijen, M. et al. Ductal carcinoma in situ: to treat or not to treat, that is the question. Br. J. Cancer 121, 285–292 (2019).
Elmore, J. G. et al. Diagnostic concordance among pathologists interpreting breast biopsy specimens. J. Am. Med. Assoc. 313, 1122–1132 (2015).
Simpson, P. T., Reis-Filho, J. S., Gale, T. & Lakhani, S. R. Molecular evolution of breast cancer. J. Pathol. 205, 248–254 (2005).
Ramkumar, C., Prakash, C., Madhav, L., Kumar, A. & Basavaraj, C. Assessment of Ki67 as a prognostic marker in hormone receptor positive breast cancer: a retrospective study on an Indian cohort. J. Mol. Biomark. Diagn. (2017).
Balaur, E., Sadatnajafi, C. & Abbey, B. Large-scale fabrication of optically active plasmonic arrays via displacement Talbot lithography. J. Phys. Conf. Ser. 1455, 012005 (2020).
Balaur, E., Sadatnajafi, C., Kou, S. S., Lin, J. & Abbey, B. Continuously tunable, polarization controlled, colour palette produced from nanoscale plasmonic pixels. Sci. Rep. 6, 28062 (2016).
Balaur, E., Sadatnajafi, C., Langley, D. & Abbey, B. Optimisation of polarization controlled colour tuning using nanoscale cross-shaped apertures in silver films. Proc. SPIE 10013, 100132F (2016).
Namatsu, H. et al. Three-dimensional siloxane resist for the formation of nanopatterns with minimum linewidth fluctuations. J. Vac. Sci. Technol. B 16, 69 (1998).
Giannuzzi, L. A. Introduction to Focused Ion Beams 1st edn (Springer US, 2005).
Jung, L. S., Campbell, C. T., Chinowsky, T. M., Mar, M. N. & Yee, S. S., Quantitative interpretation of the response of surface plasmon resonance sensors to adsorbed films. Langmuir 14, 5636–5648 (1998).
Maier, S. A. Plasmonics: Fundamentals and Applications (Springer US, 2007).
Balaur, E. et al. Plasmon-induced enhancement of ptychographic phase microscopy via sub-surface nanoaperture arrays. Nat. Photon. 15, 222–229 (2021).
Sadatnajafi, C., Balaur, E. & Abbey, B. Bimodal plasmonic color filters enable direct optical imaging of ion implantation in thin films. Adv. Funct. Mater. https://doi.org/10.1002/adfm.202009419 (2021).
Simmons, D. M. & Swanson, L. W. Comparing histological data from different brains: sources of error and strategies for minimizing them. Brain Res. Rev. 60, 349–367 (2009).
Niazi, M. K. K. et al. Relationship between the Ki67 index and its area based approximation in breast cancer. BMC Cancer 18, 867 (2018).
Davie, S. A. et al. Effects of FVB/NJ and C57Bl/6J strain backgrounds on mammary tumor phenotype in inducible nitric oxide synthase deficient mice. Transgenic Res. 16, 193–201 (2007).
Orian, J. M. et al. Insertional mutagenesis inducing hypomyelination in transgenic mice. J. Neurosci. Res. 39, 604–612 (1994).
Matenaers, C., Popper, B., Rieger, A., Wanke, R. & Blutke, A. Practicable methods for histological section thickness measurement in quantitative stereological analyses. PLoS ONE 13, e0192879 (2018).
Thurby, C., White, M. R., Simutis, F., Gillham, S. & Slinker, D. An improved method of tissue adhesion on glass microslides for immunohistochemical evaluation by the use of selected tissues from the dog and monkey. J. Histotechnol. 32, 198–201 (2009).
Cruz-Roa, A. et al. Accurate and reproducible invasive breast cancer detection in whole-slide images: a deep learning approach for quantifying tumor extent. Sci. Rep. 7, 46450 (2017).
B.A., E.B. and K.A.N. acknowledge the support of the Australian Research Council through the Centre of Excellence in Advanced Molecular Imaging (CE140100011). Fellowship funding from the Victorian Cancer Agency and grant funding from the National Breast Cancer Foundation NBCF (IIRS-21-069) is acknowledged by B.S.P. Funding from the National Breast Cancer Foundation (NBCF Prac. 16-006) and Sydney Breast Cancer Foundation is acknowledged by S.O’T. B.A. acknowledges support from the La Trobe Biomedical and Environmental Sensor Technology (BEST) Research Centre. The authors gratefully acknowledge X. Li from the La Trobe Statistics Consultancy Platform for help and advice with the statistical analysis of the data. This work was performed in part at the Melbourne Centre for Nanofabrication (MCN) in the Victorian Node of the Australian National Fabrication Facility (ANFF).
The authors declare no competing interests.
Peer review information Nature thanks Roberto Salgado and the other, anonymous, reviewer(s) for their contribution to the peer review of this work.
Publisher’s note Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Extended data figures and tables
a, FEM model of a section of the nanoslide showing the glass substrate (dark blue), nanoaperture in the metallic film, the capping layer, and air (light blue) constructed in the COMSOL Multiphysics software program. Simulated transmission spectra from each of the four lines (I-IV) which make up the chevron sample (see Fig. 2) and the background (B) spectrum generated using b, FEM model of the device for circular apertures (diameter = 160 nm). c, Shows the simulated background spectrum (no sample) for the crosses (arm length = 160 nm), ellipse (major axis = 210 nm, minor axis = 160 nm), and circular apertures. The peaks represent the plasmon resonances that occur within the active layer of the device. d, Example FEM simulations of a cross section of the electric field generated by a single circular nanoaperture at the peak resonant wavelength of the spectra shown in b
a, Schematics of the chevron structure and b, the ‘staircase’ sample. c, The AFM line-scan height profile of the chevron structure sample, the thinnest sample stripe is 3 ± 1 nm. d, The AFM line-scan height profile across the steps and stripes respectively of the ‘staircase’ sample
Extended Data Fig. 3 Pathology workflow for small-animal study and schematic showing serial sections used.
a, Pathology workflow for small-animal, MMTV-PyMT mouse model study. b, Schematic showing how serial sections were taken to enable a direct comparison of nanoslide, H&E, and Ki67.
The Ki67 and nanoslide positive areas are overlaid. Large-field-of-view (3.8 × 3.8 mm2) areas of tissue stained with H&E and Ki67 compared to the results obtained directly on nanoslide. The areas of positivity of both nanoslide (light blue) and Ki67 (bright green) were identified based on their respective HSL colour space values for example, ref. 32 (also see Methods).
Bright-field optical images: H&E stained images and histoplasmonic images. Tissue slices were placed on the slides in a sequential manner.
a, Histology images (200× magnification) were sub-categorized into four different stages for both nanoslide (1st column) and Ki67 (3rd column). The HSL image pixel colour space values were compared against ground truth pathology annotations and classified as True Positive (TP - green), False Negative (FN - red), False Positive (FP - yellow), and True Negative (TN - blue). The white space in the Ki67 image in the top row is an area where no stain adhered. Scale bars = 15 μm. b, H&E images for neoplastic regions –yellow outline (1st column), nanoslide intensity (2nd column), and Ki67 (3rd column) positivity.
Example histoplasmonic images of 1 μm thick sequential cancerous breast tissues (PyMT mice). a, Low magnification ( × 100) images. Left: contouring of cancerous regions on Nanoslide. Middle: contouring of the same cancerous regions using Ki67. Right: the contours for Nanoslide (blue) is overlapped with the contour for Ki67 (red). b, High magnification ( × 200) images. Left: Nanoslide image of MIN region; Middle: Ki67 image of the same region. Right: Overlap of positive cells on Nanoslide and MIN (92% concordance). c, High magnification (×200) images. Left: Nanoslide image of MIN region; Middle: Ki67 image of the same region. Right: Overlap of positive cells on Nanoslide and MIN (93% concordance).
Schematic showing how serial sections were taken to enable a direct comparison of nanoslide, H&E, ER, and CK 5/6 for human tissue.
Extended Data Fig. 9 Bright-field microscopy of 70 nm thick TEM optic nerve tissue sections imaged using toluidine staining and with histoplasmonics.
Top: Bright-field optical image of toluidine stained 70 nm thick optic nerve sections. Nanoslide results: middle – at 0° incident polarization, bottom – at 90° incident polarization reveals axons, glia and the myelin sheath as different colours. The black scale bar in both the left and right column images is 5 μm.
Left: Histoplasmonic images of cancerous breast tissues (PyMT mice) sectioned at 4 and 5 μm slice thickness showing almost no difference in colour contrast. Right: Bright-field optical images of optic nerve sections sectioned at 70 and 200 nm thickness showing dramatic difference in colour contrast. The scale bar is 15 μm.
This file contains Supplementary Figures 1-6 and Supplementary Tables 1-4.
Example video of pathologist using nanoslide for histoplasmonics. Video shows an example of breast tissue being examined on nanoslide. Note that the colours as they appear in the video look slightly different to when images are captured directly using the microscope.
About this article
Cite this article
Balaur, E., O’ Toole, S., Spurling, A.J. et al. Colorimetric histology using plasmonically active microscope slides. Nature 598, 65–71 (2021). https://doi.org/10.1038/s41586-021-03835-2